The emergence of various faults are because of improper operation may cause the device or the device itself when users are in the operation of salt spray corrosion testing Machine , do not worry! Here in specifically for salt spray test machine we summarize the various failures of easy method, view a list of easy to look over, to deal with problem .salt spray test machine has heat, salt spray, mildew use, artificial climate. Widely used in electronic, electrical and automotive, motorcycle, hardware tools and other products, components, materials and products, metal plating and coating salt spray corrosion test.
Failure 1: Not function properly after the power is turned
Troubleshooting methods: check the water level inside the heat sink is too low, when the water level is too low when the device will cut off the power supply to ensure safe operation of the equipment; therefore time when such a failure occurs, first check the heat sink The water level, if such a failure is still not exhaust the normal operation, this time must be the salt spray test equipment functional failure occurs, you can contact us,Haida Equipment Co., Ltd. http://www.qc-test.com .
Failure 2: Salt spray test chassis temperature can not rise to the set value
Troubleshooting: 1, if the laboratory temperature controller temperature set too low, we can set the temperature controller at the desired temperature. 2, if the laboratory safety protection switch set too low, we can be a safety switch set to the desired temperature. 3, if the heating system, electromagnetic relay, controller fails, you can contactDongguan Haida Equipment Co., Ltd . http://www.qc-test.com .
Fault 3: Normal spray and air compressor did not operate normally
Troubleshooting Methods: Under normal circumstances, for safety considerations equipment, air compressors salt spray test machine is generally equipped with automatic protection system, when there is a non-normal use salt spray test machine, the compressor will automatically stop running, so When such a failure occurs, you can be used as usual.
Troubleshooting 4:Salt spray test machine equipment spray untimed
Troubleshooting: When the occurrence of such failure, the laboratory should immediately check whether the temperature rises to a specified temperature, because if the temperature does not reach the preset temperature, timing equipment must think the experiment has not yet started, so such failures can check the preset temperature and the actual temperature can be discharged; course, there may be set for the same time and spraying time, making it impossible to observe the timing of time.
Failure 5: Touch screen does not respond
Troubleshooting Methods: Since salt spray test equipment gradually intelligent, so now most of the factory's salt spray test machine with touch screen; screen lock after exclusion factors, the instrument can contact Dongguan Haida Equipment Co., Ltd .
Failure 6: Saturated salt spray test machine barrel can not rise to the set value
Troubleshooting: 1, if the temperature is too low saturation barrel temperature controller, we can set the temperature control to the desired temperature. 2, if the saturation barrels security switch is set too low, we can protect the safety switch to the desired temperature. 3, if the heating system, electromagnetic relay, controller fails, you can contact Dongguan Haida Equipment Co., Ltd. and repair of equipment salt spray corrosion test machine after-sales department.
Fault 7: Insufficient amount of salt spray spray machine
Troubleshooting: 1, if the spray regulator placed too low, we can increase the spray regulator. 2, if the tank is preheated glass filter clogging, it may be that we can clean glass filter. 3, if the pressure is set too low, the regulator may be adjusted 1KG we per square centimeter pressure, a compressor regulating valve marked adjusted to 2KG per square centimeter pressure. 4, if the machine water shortage warning lights, which means that the water level is too low, you need to check whether the lab or saturated enough water bucket. 5, if the compressor is not running, because the air compressor itself has self-reported function, this can be used as usual, does not affect the testing process tests. 6, if the tank is not running, the water level is too low can cause heat sink off the power, you can heat the water level inside the tank under normal conditions can be added to. 7, if the temperature monitors or temperature probe fails, you can contact Dongguan Haida Equipment Co., Ltd. and repair of equipment JOOWAY salt spray corrosion test machine after-sales department.
Fault 8: Salt spray test machine Salt settlement amount is too small
Troubleshooting: 1, adjust salt spray tower at the top of the cap and spray tip from the mouth, the greater the distance the greater the amount of the settlement. 2, spray pressure regulating valves, pressure valves larger the greater the MPA Salt settlement, the settlement amount is proportional to the MPA and salt spray.
We are here to help you! If you close the chatbox, you will automatically receive a response from us via email. Please be sure to leave your contact details so that we can better assist